Power Measurement in the Lucent360™
To measure the effective irradiation of samples in the Lucent360™, we performed actinometric experiments with several vial configurations. Actinometry is the ideal method to determine how many photons penetrate a reaction vial. The results depend on the geometry of the vial, the reaction volume and the position relative to the light sources. We use a solution of ferrioxalate which is sensitive to light from 365 nm to 475 nm. The Fe3+ is reduced to Fe2+ . The amount of produced Fe2+ is proportional
to the number of photons absorbed by the solution. It is possible to determine the photon flux absorbed by an actinometric solution. You can learn more about actinometry here:
In the Lucent360™ we performed several preliminary measurements, and we determined this effective power and photon flux.
See below measurements at 450 nm.
|Volume||Vial||Measured Effective Power||Measured Photon Flux||Light Configuration|
|2 ml||4 ml||1 W||4×10-6 Einsten/s|
|10 ml||20 ml||5 W||19×10-6 Einsten/s|
|100 ml||700 ml (Reactor)||40 W||150×10-6 Einsten/s|
For reference the irradiation in a PhotoRedOx Box reaches 0.35W for 2 ml into a 4 ml vial.
More actinometry results will be disclosed in the coming weeks.
Interested in learning more about our products?
Complete our short contact form and we'll get back to you as soon as possible.